Digital Pressure module, 0.02%FS, 0 to 15 psi, gauge pressure, 1/2NPT M
$1,713.18
Digital Pressure module, 0.02%FS, 0 to 15 psi, gauge pressure, 1/2NPT M
MANUFACTURER: ADDITEL
PART NO: ADT160A-02-GP15-PSI-N2
- Description
- Size Guide
- SPECIFICATION
- DATASHEET
Description
Digital Pressure module, 0.02%FS, 0 to 15 psi, gauge pressure, 1/2NPT M
With advanced microprocessor technology and state-of-the-art silicon pressure sensors, the Additel 160 series Intelligent Digital Pressure Module (IDPM) provides an accurate, reliable, and economic solution for wide range of pressure applications. It is loaded with functionality and remarkably easy to use. In order to reach the best performance, every silicon pressure sensor in the module has been specially aged, tested and screened before assembly. Designed as external pressure module for Additel 760 automatic handheld pressure calibrator, Additel 761 automated pressure calibrator and Additel 222A, Additel 223A and Additel 761 calibrators, the Additel 160 IDPM is unmatched in performance and reliability.
K | 40000 | 2800 | G , L | 0.1 | 1.35x |
GP50K | 50000 | 3500 | G , L | 0.1 | 1.2x |
GP60K | 60000 | 4200 | G , L | 0.1 | 1.1x |
[1]. Sealed gauge pressure for above 1000 psi [2]. G=Gas, L=Liquid |
Differential Pressure | ||||||
---|---|---|---|---|---|---|
P/N | Pressure Range[1] | Media | Accuracy (%FS) | Burst Pressure | Static Pressure Range | |
(inH2O) | (mbar) | |||||
DP1 | ±1 | ±2.5 | G | 0.05[2] | 100× | ±10 psi |
DP2 | ±2 | ±5.0 | G | 0.05[2] | 100× | ±10 psi |
DP5 | ±5 | ±10 | G | 0.05[2] | 50× | ±10 psi |
DP10 | ±10 | ±25 | G | 0.05[2] | 20× | ±10 psi |
DP20 | ±20 | ±50 | G | 0.05 | 20× | ±10 psi |
DP30 | ±30 | ±75 | G | 0.05 | 20× | ±10 psi |
DP50 | ±50 | ±160 | G | 0.05 | 3× | ±10 psi |
DP150 | ±150 | ±350 | G | 0.02 | 3× | 50 psi |
DP300 | ±300 | ±700 | G | 0.02 | 3× | 50 psi |
[1] FS specification applies to the span of the range. Accuracy includes 1 year stability. [2] 0.05%FS accuracy (incl 6 months stability). One year accuracy is 0.05%FS calibration accuracy combined with 0.05%FS one year stability. |
Compound Pressure | ||||||
---|---|---|---|---|---|---|
P/N | Pressure Range | Media | Accuracy | Pressure Rating | ||
psig | bar.g | Burst | Over Pressure | |||
CP2 | ±2 | ±0.16 | G | 0.05% FS | 3x | 1.2x |
CP5 | ±5 | ±0.35 | G | 0.02% FS | 3x | 1.2x |
CP10 | ±10 | ±0.7 | G | 0.02% FS | 3x | 1.2x |
CP15 | ±15 | ±1.0 | G | 0.02% FS | 3x | 1.2x |
CP30 | -15 to 30 | -1 to 2.0 | G | 0.02% FS | 3x | 1.2x |
CP50 | -15 to 50 | -1 to 3.5 | G | 0.02% FS | 3x | 1.2x |
CP100 | -15 to 100 | -1 to 7.0 | G,L | 0.02% FS | 3x | 1.2x |
CP300 | -15 to 300 | -1 to 20 | G,L | 0.02% FS | 3x | 1.2x |
CP500 | -15 to 500 | -1 to 35 | G,L | 0.02% FS | 3x | 1.2x |
CP600 | -15 to 600 | -1 to 40 | G,L | 0.02% FS | 3x | 1.2x |
CP1K | -15 to 1,000 | -1 to 70 | G,L | 0.02% FS | 3x | 1.2x |
CP2K | -15 to 2,000 | -1 to 140 | G,L | 0.02% FS | 3x | 1.2x |
CP3K | -15 to 3,000 | -1 to 200 | G,L | 0.02% FS | 3x | 1.2x |
CP5K | -15 to 5,000 | -1 to 350 | G,L | 0.02% FS | 3x | 1.2x |
CP10K | -15 to 10,000 | -1 to 700 | G,L | 0.02% FS | 2x | 1.2x |
Absolute Pressure | |||||
---|---|---|---|---|---|
P/N | Pressure Range | Media | Accuracy (%FS) | Burst Pressure | |
psi | bar | ||||
AP5 | 5 | 0.35 | G | 0.1 | 3× |
AP10 | 10 | 0.7 | G | 0.1 | 3× |
AP15 | 15 | 1 | G | 0.1 | 3× |
AP30 | 30 | 2 | G | 0.1 | 3× |
AP50 | 50 | 3.5 | G | 0.1 | 3× |
AP100 | 100 | 7 | G , L | 0.05 (0.1) | 3× |
AP300 | 300 | 20 | G , L | 0.05 (0.1) | 3× |
AP500 | 500 | 35 | G , L | 0.05 (0.1) | 3× |
AP1K | 1000 | 70 | G , L | 0.05 (0.1) | 3× |
AP3K | 3000 | 200 | G , L | 0.05 (0.1) | 3× |
AP5K | 5000 | 350 | G , L | 0.05 (0.1) | 3× |
Precision Sensors[1] – Absolute Pressure | ||||||
---|---|---|---|---|---|---|
P/N | Pressure Range | Media | Accuracy[2] | Pressure Rating | ||
psia | bar.a | Burst | Over Pressure | |||
AP15R | 0 to 15 | 0 to 1 | G | 0.01% FS | 2x | 1.2x |
AP30R | 0 to 30 | 0 to 2 | G | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP50R | 0 to 50 | 0 to 3.5 | G | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP100R | 0 to 115 | 0 to 8 | G,L | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP300R | 0 to 315 | 0 to 21 | G,L | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP500R | 0 to 515 | 0 to 36 | G,L | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP1KR | 0 to 1,015 | 0 to 71 | G,L | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP2KQ | 0 to 2,015 | 0 to 141 | G | 0.005% rdg + 0.005% FS | 3x | 1.1x |
AP3KQ | 0 to 3,015 | 0 to 201 | G | 0.005% rdg + 0.005% FS | 3x | 1.1x |
AP6KQ | 0 to 6,000 | 0 to 400 | G,L | 0.01% FS | 3x | 1.2x |
AP10KQ | 0 to 10,000 | 0 to 700 | G,L | 0.01% FS | 2x | 1.2x |
AP15KQ | 0 to 15,000 | 0 to 1000 | G,L | 0.01% FS | 2x | 1.2x |
AP20KQ | 0 to 20,000 | 0 to 1400 | G,L | 0.01% FS | 2x | 1.2x |
AP30KQ | 0 to 30,000 | 0 to 2000 | G,L | 0.02% FS | 1.5x | 1.1x |
AP40KQ | 0 to 40,000 | 0 to 2800 | G,L | 0.02% FS | 1.5x | 1.1x |
[1] Contact Additel for other range options. [2] Accuracy includes calibration uncertainty, linearity and long-term stability. |
Barometric Pressure | |||||
---|---|---|---|---|---|
P/N | Pressure Range | Media | Accuracy | Burst Pressure | |
Low | High | ||||
BP | 60 kPa | 110 kPa | G | 40 Pa | 3× |